Retinal prosthesis with a new configuration
Polymer materials are useful as electrode array bodies for neural stimulation. They are particularly useful for retinal stimulation to create artificial vision, cochlear stimulation to create artificial hearing, and cortical stimulation, and many related purposes. The pressure applied against the retina, or other neural tissue, by an electrode array is critical. Too little pressure causes increased electrical resistance, along with electric field dispersion. Too much pressure may block blood flow. Common flexible circuit fabrication techniques generally require that a flexible circuit electrode array be made flat. Since neural tissue is almost never flat, a flat array will necessarily apply uneven pressure. Further, the edges of a flexible circuit polymer array may be sharp and cut the delicate neural tissue. By applying the right amount of heat to a completed array, a curve can be induced. With a thermoplastic polymer it may be further advantageous to repeatedly heat the flexible circuit in multiple molds, each with a decreasing radius. Further, it is advantageous to add material along the edges. It is further advantageous to provide a fold or twist in the flexible circuit array. Additional material may be added inside and outside the fold to promote a good seal with tissue.
Second Sight Medical Products, Inc.
H H H H H H Y H H H H H H H A A A
H05K H05K H05K H05K H05K H05K Y10T H05K H05K H05K H05K H05K H05K H05K A61N A61N A61N
additional additional additional additional additional inventional additional additional additional inventional additional inventional additional additional inventional inventional inventional
Jordan M. Neil Hamilton Brian V. Matthew J. Robert J. Kelly H. James Singleton
Neysmith Talbot Mech McMahon Greenberg McClure Little